- Open Access
Plant cell wall characterization using scanning probe microscopy techniques
© Yarbrough et al; licensee BioMed Central Ltd. 2009
- Received: 6 December 2008
- Accepted: 24 August 2009
- Published: 24 August 2009
Lignocellulosic biomass is today considered a promising renewable resource for bioenergy production. A combined chemical and biological process is currently under consideration for the conversion of polysaccharides from plant cell wall materials, mainly cellulose and hemicelluloses, to simple sugars that can be fermented to biofuels. Native plant cellulose forms nanometer-scale microfibrils that are embedded in a polymeric network of hemicelluloses, pectins, and lignins; this explains, in part, the recalcitrance of biomass to deconstruction. The chemical and structural characteristics of these plant cell wall constituents remain largely unknown today. Scanning probe microscopy techniques, particularly atomic force microscopy and its application in characterizing plant cell wall structure, are reviewed here. We also further discuss future developments based on scanning probe microscopy techniques that combine linear and nonlinear optical techniques to characterize plant cell wall nanometer-scale structures, specifically apertureless near-field scanning optical microscopy and coherent anti-Stokes Raman scattering microscopy.
- Atomic Force Microscopy
- Plant Cell Wall
- Stimulate Raman Scattering
- Scanning Probe Microscopy
- Parenchyma Cell Wall
Lignocellulosic biofuels have been widely proposed as the most promising route for the sustainable and renewable production of liquid transportation fuels today . There are millions of tons of lignocellulosic biomass, such as wood chips, agricultural residues, and grasses that could be converted to simple sugars and then fermented to transportation fuels. One key bottleneck for the current bioconversion technologies used to produce biofuels is the high cost and low yield of fermentable sugars (mainly glucose and xylose) derived from plant biomass. The cell walls in plant biomass are composed primarily of polysaccharides and a non-fermentable polyphenolic fraction, lignin. Cellulose is the major plant cell wall polysaccharide and consists entirely of non-branched β-1,4-linked glucose. The hemicelluloses comprise a variety of branched and esterified polysaccharides containing xylose, mannose, arabinose, galactose, and glucose linked by various glycoside bonds . The dominant hemicellulose in grasses and hard woods, the main energy feedstocks, is arabinoxylan. Pectins are defined by the presence of uronic acids as major components . Although lignin biosynthesis is less understood than the biosynthesis of polysaccharides in plant cell walls, it has been established that the composition of lignin in higher plants is derived from three monolignols: p-coumaryl, p-coniferyl, and E-sinapyl alcohol. These components are thought to play a vital role in the function and stability of the plant cell wall and constitute, in part, it's recalcitrant nature .
In concert, these biopolymers form the complex networks comprising the plant cell wall, where cellulose constitutes the core of the relatively rigid microfibrils and other cell wall polymers form 'matrixing' materials binding the system into a kind of 'liquid polymer crystal'. To deconstruct the plant cell wall and liberate these mixed sugars efficiently, the insoluble structural carbohydrates present must be first available to, and then converted to soluble sugars by cellulolytic and hemicellulolytic enzymes. In order to enhance enzyme digestibility, current biomass conversion technologies use thermaland/or chemical pretreatment to make the cell walls more amenable to these enzymes, although the detailed mechanisms of how these enzymes act on the complex plant cell wall substrate remains poorly understood.
As the primary structural component of the plant cell wall, cellulose forms the cores of plant cell wall microfibrils. The biophysical properties of cellulose are based almost entirely on how these glucan chains are packed (that is, cellulose can form the polymorphs Iα, Iβ, II, III and IV), which vary depending on the size of the fiber, the fiber length, the chain angle, and the solvent in which it is studied. Nevertheless, the structures of cellulose reported in the literature are primarily based on either highly modified or regenerated plant cellulose, or large cellulose crystals produced by algae or animals. The native structure of plant cell wall cellulose remains unknown primarily because insufficient tools are available to study cellulose microfibrils (2 to 10 nm in diameter) in situ or in vivo [4–6]. Even the number of cellulose chains in the elementary fibril is not known , although most previous work has proposed a 36-chain model based primarily on a single cellulose synthase complex or 'rosette' containing 36 subunit enzymes. There is evidence that cellulose microfibril formation is under dynamic cellular control regulated, at least in part, by the density and arrangement of the synthetic sites, as well as the identity and availability of other cell wall polymers that can co-aggregate with the forming cellulose fibrils . Cellulose fibrils form an essential scaffold for many plant cell walls and their absence disrupts the hierarchical assembly of other cell wall components , such as hemicelluloses, pectins, and even lignins. Readers can find comprehensive recent reviews about plant cell wall synthesis elsewhere [3, 6, 9].
Plant cell wall structural elements are on the order of nanometers and, unfortunately, traditional optical microscopy does not allow direct observation of these materials. It is now clear that a better fundamental understanding of how these materials assemble to form the plant cell wall is critical. In order to accomplish this objective, characterization of these structures with regard to their chemical and physical features at the nanometer scale is necessary. Unfortunately, very few microscopy techniques permit direct measurement at this high spatial resolution without extensive and time-intensive sample preparation that can potentially damage or alter the plant material. Traditionally, these measurements have been made with scanning electron microscopy (SEM) and transmission electron microscopy (TEM). Sample preparation for TEM involves the embedding of the plant specimens into a resin for stabilization and then thin sectioning to a suitable thickness. For high-resolution SEM the sample must be made conductive, which is normally accomplished by coating biological specimens with vaporized metal or carbon. It is believed that these sample preparation procedures used for the plant materials can damage and ultimately change the native structure of the plant cell wall. With the recent introduction of high-pressure freezing and freeze substitution methods for preserving cells for ultrastructural analysis, electron microscopy has become a valuable tool for correlative imaging. However, there are still limitations with these imaging techniques and often the reduction of resolution is the resulting compromise [10–13].
To this end, scanning probe microscopy (SPM) provides a critical imaging modality that offers the ability to investigate these materials at the nanometer scale with essentially no sample preparation or perturbation, using a distinctive 'biophysical' methodology. In this context, we will review the principles of SPM and its application to imaging plant cell walls and cellulose crystals. Furthermore, we provide a discussion regarding the possible benefits of combing SPM with tip-enhanced near-field spectroscopy, as well as discussing potential correlations between SPM and traditional electron microscopy.
Scanning probe microscopy and its application to imaging cellulose and the plant cell wall microfibril
SPM utilizes the ability to measure the interaction between a fine physical probe and the surface of the sample. Atomic force microscopy (AFM) and near-field scanning optical microscopy (NSOM) are two examples of SPM techniques that can obtain a spatial resolution at the sub-micron level (that is, between 0.1 nm and 100 nm). These techniques permit the direct characterization of the surface of biological systems with high spatial resolution and minimal sample preparation. They are ideal tools for characterizing the ultrastructure and biochemistry of the plant cell wall because many critical features of the cell wall lie within their detection range.
AFM is typically used to measure the interaction forces between the probe tip's radius of curvature and the sample surface. The tip radius is typically 5 to 10 nm, but can be as small as 1 nm. These tips are usually made from silicon or silicon nitride. AFM systems typically use an optical feedback circuit whereby a laser spot is reflected off the cantilever holding the tip and the deflection of the tip is measured as the tip is rastered across the sample surface. AFM can be used to measure primarily the physical or topographical, as well as some chemical, properties of the sample surface. NSOM; however, is typically used to measure the optical (and thus chemical) properties of materials in registry with their topography. This is accomplished by utilizing a tapered optical fiber as the probe. Since an optical fiber probe is used, NSOM has the ability to optically characterize (with wavelengths in the visible spectrum) a sample on the submicron level by permitting simultaneous injection of light into, or collection of light from the sample. This ability allows the optical properties of the sample, fluorescently tagged labels, and other optical techniques to be mapped in registry with topography at the nanometer scale.
All tip-based scanning probe microscopy uses an auxiliary gap-width sensing mechanism (for example, mechanical shear force, optical detection) in a feedback loop allowing a fixed tip-sample separation to be maintained while rastering the probe along the sample surface. This method allows scanning with small tip-sample separations without damaging the tip or the sample. Unfortunately, this type of scanning mechanism can give rise to artifacts, arising from cross talk between topographic and other signals caused by the gap-width control mechanism.
Along with the introduction of artifacts associated with a moving tip, there are other challenges associated with SPM. For example, as it is the surface of the material being probed, there is no direct measurement of the bulk properties of the material. Also, in most probe-type microscopy experiments there is a tendency for the probe tip to physically disturb molecules (especially biomolecules) that are not firmly attached to the substrate surface . Finally, hysteresis plays a vital role in the repeatability of scanning probe microscopy, because it can lead to image distortions and drift of the sample . With regard to both AFM and NSOM, steps can and should be taken to aid in determining whether or not tip artifacts influence the images. Considerable literature exists addressing this subject and it is not within the scope of this article to delve into greater detail about the techniques used for determining the presence of tip artifacts and strategies for overcoming these artifacts. Instead, the reader is directed to several articles that offer greater detail on this subject [16–26].
Kirby et al.  performed AFM measurements on several different plant cell wall materials that were taken from homogenized Cox orange pippin apples, water chestnuts, Bintji potato, and Amsterdamse bak carrot. Their work demonstrated the feasibility of using AFM to probe the molecular architecture of hydrated plant cell walls and also showed that sample preparation (in this case, freeze-thawing) could result in minor differences in the structure before and after treatment . Their measurements showed clear evidence that the cell wall consisted of a laminated structure with different layers of fibers orientated in different directions, which supported the general assumption that typical plant cell walls are polylaminate structures. Kirby et al. also demonstrated that the isolated fibers found in the top layer were thicker than fibers that were ordered into aligned arrays or layers. This apparent thickness difference between fibers was attributed to a probe-broadening artifact .
Davis and Harris  performed AFM measurements on cell walls from onion and Arabidopsis thaliana. Their measurements showed tightly interwoven microfibrils, which in both species were approximately the same width. In addition to these measurements, they went on to determine microfibril diameters in samples that had been subjected to different treatment conditions. It was found that removal of pectic polysaccharides could improve the accuracy of measurements of the microfibril dimensions. These treatments consisted of using either a combination of trans-1,2-diamino-cyclohexane N, N, N', N'-tetraacetic acid (CDTA) with sodium carbonate or KOH. It was also reported from nuclear magnetic resonance (NMR) studies that the CDTA mixture did not affect the size or crystallinity of cellulose, whereas the KOH solution did alter the ordering of the cellulose and resulted in the production of both cellulose II and amorphous cellulose, along with possible swelling of the microfibrils. Different diameters of microfibrils were reported from different plant materials, which ranged 4.4 to 7.9 nm depending on the treatment used. Interestingly, these microfibrillar diameters were smaller than those reported previously (~25 nm) by Kirby et al. , the latter of which could be attributed to tip-broadening effects. Kirby and co-workers then suggested the use of height measurements to avoid the potential probe-broadening artifacts. Davis and Harris  also investigated the tip-broadening problem by performing multiple scans at different scales (magnifications). They were able to conclude that tip artifacts were unlikely contributors to the contours of the microfibrils observed, because they were not consistent throughout the image and they were seen in other scans of the same sample.
We recently reported [27, 28] AFM imaging of the maize parenchyma cell wall surface. In this study, mature parenchyma cell walls from naturally senescent stem pith were imaged without any chemical treatment. The thick fibers that appeared to exist only in the uppermost surface of some parenchyma cell walls were also observed, which were consistent with the observations of Kirby et al. . These structures were not found in cell walls in which the microfibrils appeared to be heavily coated with matrices suspected of containing hemicelluloses and pectins. We found that the larger fibers observed in the top layer of the cell wall appeared to be bundles of smaller fibers, which did not appear to result from tip-broadening as interpreted by Kirby et al. . We maintained that they were bundles of newly synthesized cellulose elementary fibers (CEFs), which we later termed 'macrofibrils'. We also reported seeing similar random orientation of the microfibrils reported by Davis and Harris . From these studies, we further reported the diameter of the CEF was 3 to 5 nm measured using AFM height imaging, which was smaller than that measured by previous AFM studies. This measured diameter seemed consistent with the general model of a 36-chain elementary fibril . These models suggested that the elementary fibril is synthesized from one cell wall 'rosette' containing 36 cellulose synthase enzymes (CesA). As with the other AFM measurements reported here, we performed additional tests to eliminate the possibility of tip artifacts. These procedures included calibration of each tip with a reference gold particle sample, interchanging of tips, and a consistent scan-size strategy to ensure reproducibility.
AFM is a powerful tool that can provide high topographic resolution. However, one of the limitations of AFM is its poor chemical resolution capability, especially for imaging complex biomaterials, such as plant cell walls. As previously stated, these plant materials could be used for the production of lignocellulosic biofuels; however, the greatest barrier for commercial processes is overcoming the natural resistance of cell walls to deconstruction. To accomplish this task in a cost-effective manner, a better understanding of the fundamental chemical structure of the plant cell wall as well as the overall effects of pretreatment is needed.
Traditional microscopy and spectroscopy techniques can readily characterize these plant materials in bulk form (for example, at the macroscopic scale). Unfortunately, these diffraction-limited spectroscopic techniques are proving to be a barrier to advancing our fundamental understanding of these materials. It is now clear that in order to study the fundamental structure of cell walls at the nanometer scale, more advanced scanning probe techniques are needed. Ideally, such techniques would combine diverse spectroscopic techniques, including fluorescence, Raman scattering, and nonlinear optical processes, such as two-photon fluorescence, second harmonic generation, and third harmonic generation. With this objective in mind, recent developments in combining SPM techniques and linear or non-linear spectroscopy appear promising for achieving the required high spatial and chemical resolution, simultaneously.
High optical and/or chemical and spatial resolution can be achieved in NSOM using a tapered optical fiber with a sub-wavelength aperture. This sub-wavelength aperture (approximately 100 nm) is formed at the end of the tapered optical fiber by evaporating aluminum on all sides of the tapered sides of the optical fiber. There are several other methods to form this sub-wavelength aperture and each method has its advantages and disadvantages (for detailed methods, see [21, 34]. However, smaller and larger apertures can be obtained and the size of the aperture typically governs the optical resolution, as well as the topographical resolution.
Optical microscopy is traditionally used to collect these spectroscopic signals in the far field, or to be more precise, above the diffraction limit of light where light begins to interfere with itself. As these systems operate in the far-field zone, they all have a spatial resolution of approximately λ/2 (that is, the Rayleigh criterion). This situation prohibits our ability to optically resolve and/or distinguish features smaller than the diffraction limit. This objective is a big challenge when optical microscopy is used to characterize plant cell wall constituents that are a few nanometers in size. Therefore, the signals collected by optical microscopy represent an ensemble average composed of signals from many different constituents in the plant cell wall. To increase spatial resolution, these optical techniques and SPM can be combined, which then permits measurements to be made below the diffraction barrier (within the near-field zone). These measurements could be performed on a sub-wavelength scale (that is, <<λ) where the spatial resolution is limited only by the size of the scanning probe. Recent achievements in understanding the interaction of light beyond the diffraction limit has led to an explosion of interest in using apertureless SPM to collect optical information beyond the diffraction limit of light .
The typical aperture on the end of an NSOM probe is approximately 100 nm in diameter; however, these probes can reach tip diameters as small as 20 nm. As these tips are made from an optical fiber the tip itself is very fragile and easily damaged, and this can lead to unwanted tip artifacts . Along with the fragility of the NSOM probes, their collection efficiencies are very low, typically on the order of 10-6, which results in a requirement for long integration times to collect the optical signal. Due to these limitations of the optical probe used in NSOM, the apertureless probe with tip enhancement (TE) technique is more common today for the collection of optical signals below the diffraction limit of light. NSOM based on local field enhancement was first proposed in 1985, even before the invention of the atomic force microscope . Today, this method involves using either a solid metal tip or a metalized AFM tip made by the electrochemical etching of a thin wire or by coating the tip with silver (usually a noble metal, preferably silver or gold) . The optical signal generated can be strongly enhanced by this metal coating, which significantly improves the NSOM by drastically reducing the integration time and increasing the signal-to-noise ratio . The detailed physics behind using TE-NSOM has been reviewed in elsewhere .
Considerable work is being done within this field today and these apertureless techniques are being used to perform most of the optical characterization within the near field. Fragola et al.  used apertureless near-field fluorescence microscopy on dye-doped polystyrene spheres immersed in liquid. The primary challenge they found was to distinguish the fluorescence signals from the far field (optical signal) of the polystyrene spheres and from the near field (optical component). They further used a lock-in amplifier that was synchronized to the tip oscillation frequency and demonstrated the ability to collect fluorescence in the near field only from the polystyrene spheres with high signal-to-noise ratio and high lateral resolution . Furthermore, these measurements were performed in solution, which provided a promising approach for characterizing biological samples. For the case of plant cell walls, for example, in theory NSOM could be used in a similar way to characterize the compartmentalization and distribution of cell wall matrix polymers, as well as chemically distinct microfribrils arranged in layers.
Surface-enhanced Raman scattering (SERS) has also been used to characterize organic molecules. Typically, SERS requires the deposition of a thin metallic layer onto the sample, which improves the Raman signal intensity by several orders of magnitude. This technique was developed because spontaneous Raman scattering suffers from a very low-scattering cross-section (typically, 1 in 106 photons account for the Raman signal) . Therefore, deposition of a metallic layer (typically silver) depends critically on the substrate preparation, severely limits the applicability and renders quantitative measurement almost impossible. Another problematic factor is that the deposition of a thin layer of metal on a biological sample usually destroys it . The surface-enhanced Raman techniques are thus unfavorable for characterizing biomolecules and perhaps plant cell walls, or at least must be employed with care.
Hayazawa et al.  first demonstrated tip-enhanced near-field Raman imaging using an apertureless metallic probe. Their experiment used aggregates of Rhodamine-6F and Crystal Violet molecules deposited onto a cover slip covered with an 8 mm-thick silver film. This silver film was used to enhance the surface Raman scattering effect, as well as to reduce the fluorescence background generated by energy transfer from the molecules to metal . These measurements were performed in contact (tip in contact with the sample) and non-contact modes. The Raman scattering spectra were also collected from the focused spot. Interestingly, the Raman spectra collected in contact mode contained the inherent near-field Raman scattering induced by the metallic tip, which was absent in non-contact mode. These workers also reported that the integration times were 1 second with a low level of excitation (230 μW) for each organic molecule, which demonstrated the capability of TE-NSOM Raman imaging of organic molecules. Stöckle and co-workers  also used a sharp metal tip rastered over the sample with an AFM. They performed measurements on a thin Brilliant Cresyl Blue (BCB) layer and a C60 thin film that was drop-coated onto a glass substrate using both a metalized AFM probe and an electrochemically etched gold metal wire. In both experiments, they collected the Raman spectrum with contact and non-contact modes. The Raman signal increased more than 30 times when the metalized tips were brought into contact with these thin films with a spatial resolution on the order of 50 nm.
Along with employing these apertureless techniques to perform traditional optical spectroscopy, apertureless techniques are also being used in conjunction with non-linear optical techniques, such as two-photon excitation fluorescence (TPEF). TPEF spectroscopy utilizes two photons of energy to excite a fluorophore, resulting in the emission of a fluorescent photon at a higher energy than either of the two excitatory photons. The fluorescence yield scales as E4 for TPEF, which is similar to the scaling for strong SERS . Sánchez et al.  demonstrated the feasibility of collecting near-field TPEF using a metal tip on J-aggregates of pseudoisocyanine (PIC) dye molecule suspended in polyvinyl sulfate. A mode-locked Ti:Sapphire laser providing 100 fs pulses at 830 nm was used to demonstrate a spatial resolution ranging between 15 and 30 nm for these dyes. Near-field TPEF can also be obtained using metalized AFM probes. For example, Nieman et al.  used a gold-coated AFM probe on a dried coumarin I film mounted on a cover slip that had been degreased in acetone and methanol. A mode-locked Ti:Sapphire laser with a pulsed width of approximately 200 fs at a center wavelength of approximately 790 nm was used as the excitation source for this work. Nieman et al.  also reported short integration times, strong fluorescence yield, and both an optical and topographical spatial resolution of 43.8 nm, which is well below the diffraction limit. We conclude the TE-Raman AFM imaging may have sufficient resolution to chemically characterize plant cell wall compartments such as cellular scale features, but not microfibrils.
Another widely used non-linear optical technique is coherent anti-Stokes Raman scattering (CARS). CARS is a third-order non-linear four-wave mixing optical process involving two lasers, a pump laser at a frequency of ωp and a Stokes laser at a frequency of ωs. The resulting signal from these two lasers is at the anti-Stokes frequency of 2ω p - ω s which is generated in the phase-matching directions . The CARS signal is achieved when these two lasers' beams are collinearly added and tightly focused onto a sample. This creates a composite electric field beating at the frequency ω p - ω s . If this beating frequency matches a specific vibrational Raman frequency within the molecules it causes this vibrational Raman frequency to oscillate, thereby producing a strong excitation across the whole focal volume . Recently, there has been considerable interest in performing CARS on biological samples. The benefit of CARS is it can excite specific vibrational Raman modes within the biological system when the frequency difference (ω p - ω s ) between the pump and the Stokes laser are tuned to the resonance of a specific Raman-active molecular vibration. Due to the coherent nature of the CARS possesses, CARS has higher sensitivity than spontaneous Raman microscopy.
Ichimura et al.  demonstrated the feasibility of performing near-field CARS using a metalized AFM tip on adenine molecules in a nanomeric DNA network structure, and tuned the pump and stokes frequencies to excite the 1337 cm-1 Raman mode corresponding to the adenine (ring-breathing mode of diazole). These workers observed a background CARS signal, which they attributed to the third-order non-linear susceptibility of silver (for example, due to local four-wave mixing). Noble metals used as the probe can also pose another problem besides local four-wave mixing; they can also generate a white light continuum which is induced by multi-photon excited photoluminescence due to the recombination radiation between electrons near the Fermi level and photo-excited holes in the d band . Unfortunately, the combination of these background signals competes with the CARS process. The CARS signal in this experiment largely surpassed the background signal and so the authors were able to collect a strong CARS signal. They reported that they had imaged both topographically and optically the DNA network with fast integration times (ranging between 3 to 12 minutes) and a spatial resolution of about 250 nm. The same workers  also performed near-field CARS on single-wall carbon nanotubes (SWNT) using a gold-metalized silicon AFM tip. Kawata et al.  further improved the spatial resolution to about 150 nm; in this case, the pump and stokes frequencies were tuned to excite the Raman mode corresponding to the G-band of the SWNTs at 1581 cm-1.
SPM has enhanced our ability to directly visualize the topography of plant cell wall samples at the sub-nanometer scale. These imaging techniques have led to precise measurement of the diameters of the microfibrils, and insight into the arrangement of the microfibrils in cell wall lamellae. It has been demonstrated that the chemical resolution can be significantly improved when SPM is used in conjunction with apertureless probe and near-field spectroscopy. There has been a limited amount of work reported using the apertureless near-field technique for characterizing plant cell wall materials; however, from this work we can speculate that developing the ability to simultaneously collect topological information from AFM and use it in combination with chemical information from linear or non-linear optics will provide the information required to deepen our understanding of the architecture of the plant cell wall.
These techniques, however, harbor inherent challenges that must be overcome. For example, the fundamental problems associated with the auxiliary gap-width-sensing mechanism used for AFM operation contribute to tip artifacts and misinterpretation of the data. These tip artifacts can also cause crosstalk within the optical signals, leading to a misleading optical image as well. Along with these tip artifacts, it has been shown that for apertureless techniques there are competing signals that diminish the desired signal, and thus further work is needed to overcome these unwanted effects. The next challenge in advanced imaging of the plant cell wall using SPM is the demonstration that TE-Raman, TE-CARS or TE-SRS can be used successfully to collect chemical information at the scale of the plant cell wall layers (approximately 50 to 100 nm) or even the microfibril itself (approximately 5 to 50 nm).
This work was funded by the US Department of Energy, Office of Energy Efficiency and Renewable Energy Biomass Program. The authors wish to also thank Dr Sunney Xie and colleagues at Harvard University for many helpful discussions regarding CARS, and Dr Jeff Squier and colleagues from Colorado School of Mines for many helpful discussions regarding the photon counting setup.
- Zaldivar J, Nielsen J, Olsson L: Fuel ethanol production from lignocellulose: a challenge for metabolic engineering and process integration. Appl Microbiol Biotechnol 2001, 56: 17-34. 10.1007/s002530100624View ArticleGoogle Scholar
- Donohoe B, Decker S, Tucker M, Himmel M, Vinzant T: Visualizing lignin coalescence and migration through maize cell walls following thermochemical pretreatment. Biotechnol Bioeng 2008, 101: 913-925. 10.1002/bit.21959View ArticleGoogle Scholar
- Mohnen D, Peled MB, Somerville C: Cell Wall Synthesis. In Biomass Recalcitrance: Deconstructing the Plant Cell Wall for Bioenergy. Edited by: Himmel ME. London: Blackwell Publishing; 2008:188-212.Google Scholar
- Cosgrove DJ: Wall Structure and wall loosing. A look backwards and forward. Plant Physiol 2001, 125: 131-134. 10.1104/pp.125.1.131View ArticleGoogle Scholar
- Carpita NC, Defernez M, Findlay K, Wells B, Shoue DA, Catchpole G, Wilson RH, McCann MC: Cell wall architecture of the elongating maize celeoptile. Plant Physiol 2001, 127: 551-565. 10.1104/pp.010146View ArticleGoogle Scholar
- Somerville C, Bauer S, Brinistool G, Facette M, Hamann T, Milne J, Osborn E, Paredex A, Persson S, Raab T, Vorwerk S, Youngs H: Toward a systems approach to understanding plant-cell walls. Science 2004, 306: 2206-2211. 10.1126/science.1102765View ArticleGoogle Scholar
- Doblin M, Kurek I, Jacob-Wilk D, Delmer D: Cellulose biosynthesis in plants: from genes to rosettes. Plant Cell Physiol 2002, 43: 1407-1420. 10.1093/pcp/pcf164View ArticleGoogle Scholar
- Cannon MC, Terneus K, Hall Q, Tan L, Wang Y, Wagenhart BL, Chen L, Lamport DT, Chen Y, Kieliszewski MJ: Self-assembly of the plant cell wall requires an extensin scaffold. PNAS 2008, 105: 2226-2231. 10.1073/pnas.0711980105View ArticleGoogle Scholar
- Atalla RH, Brady JW, Matthews JF, Ding S-Y, Himmel ME: Structures of Plant Cell Wall Celluloses. In Biomass Recalcitrance: Deconstructing the Plant Cell Wall for Bioenergy. Edited by: Himmel ME. London: Blackwell Publishing; 2008:94-187.Google Scholar
- McDonald KL, Auer M: High-pressure freezing, cellular tomography, and structural cell biology. Bio Techniques 2006, 41: 137-143.Google Scholar
- Giddings TH: Freeze-substitution protocols for improved visualization of membranes in high-pressure frozen samples. J Microsc 2003, 212: 53-61. 10.1046/j.1365-2818.2003.01228.xView ArticleGoogle Scholar
- Al-Amoudi A, Norlen L, Dubochet J: Cryo-electron microscopy of vitreous sections of native biological cells and tissues. J Struct Biol 2004, 148: 131-135. 10.1016/j.jsb.2004.03.010View ArticleGoogle Scholar
- Frey B, Brunner I, Walther P, Scheidegger C, Zierold K: Element localization in ultrathin cryosections of high-pressure frozen ectomycorrhizal spruce roots. 2008, 20: 929-937.Google Scholar
- Keller D, Bustamante C: Attaching molecules to surfaces for scanning probe microscopy. Biophys J 1993, 64: 896-897. 10.1016/S0006-3495(93)81449-8View ArticleGoogle Scholar
- Dirsherl K, Garnaes J, Nielsen L: Modeling the hysteresis of a scanning probe microscope. J Vac Sci Technol B 2000, 18: 621-625. 10.1116/1.591249View ArticleGoogle Scholar
- Gainutdinov RV, Arutyunov PA: Artifacts in atomic force microscopy. Russ Microelectron 2001, 30: 219-224. 10.1023/A:1011394310890View ArticleGoogle Scholar
- Kühle A, Sørensen AH, Zandbergen JB, Bohr J: Contrast artifacts in tapping tip atomic force microscopy. Appl Phys A Mater Sci & Process 1998, 66: S329-S333. 10.1007/s003390051156View ArticleGoogle Scholar
- Shiramine K, Muto S, Shibayama T, Sakaguchi N, Ichinose H, Kozaki T, Sato S, Nakata Y, Yokoyama N, Taniwaki M: Tip artifact in atomic force microscopy observations of InAs quantum dots grown in Stranski-Krastanow mode. J Appl Phys 2007, 101: 033527. 10.1063/1.2434806View ArticleGoogle Scholar
- Hecht B, Bielefeldt H, Inouye Y, Pohl DW: Facts and artifacts in near-field optical microscopy. J Appl Phys 1997, 81: 2492-2498. 10.1063/1.363956View ArticleGoogle Scholar
- Hecht B, Sick B, Wild UP, Deckert V, Zenobi R, Martin OJF, Pohl D: Scanning near-field optical microscopy with aperature probes: Fundamentals and applications. J Chem Phys 2000, 112: 7761-7774. 10.1063/1.481382View ArticleGoogle Scholar
- Pilevar S, Edinger K, Atia W, Smolyaninov I, Davis C: Focused ion-beam fabrication of fiber probes with well-defined apertures for use in near-field scanning optical microscopy. Appl Phys Lett 1998, 72: 3133-3135. 10.1063/1.121570View ArticleGoogle Scholar
- Ke SH, Uda T, Štich I, Terakura K: Effect of tip morphology on AFM images. Appl Phys A 2001, 72: S63-S66. 10.1007/s003390100639View ArticleGoogle Scholar
- Hoffmann Á, Jungk T, Soergel E: Cross-talk correction in atomic force microscopy. Rev Sci Instrum 2007, 78: 016111-0161013. 10.1063/1.2424448View ArticleGoogle Scholar
- Fujisawa S, Ogiso H: Compensation of cross talk in the optical lever deflection method used in atomic force microscopy. Rev Sci Instrum 2003, 74: 5115-5117. 10.1063/1.1622972View ArticleGoogle Scholar
- Sedin DL, Rowlen KL: Influence of tip size on AFM roughness measurements. Appl Phys A Mater Sci Process 2001, 182: 40-48.Google Scholar
- Varenberg M, Etsion I, Halperin G: Crosstalk problems in scanning-by-probe atomic force microscopy. Rev Sci Instrum 2003, 74: 3362-3367. 10.1063/1.1584082View ArticleGoogle Scholar
- Ding S-Y, Himmel ME: The maize primary cell wall microfibril: a new model derived from direct visualization. J Agric Food Chem 2006, 54: 597-606. 10.1021/jf051851zView ArticleGoogle Scholar
- Himmel ME, Ding SY, Johnson DK, Adney WS, Nimlos MR, Brady JW, Foust TD: Biomass recalcitrance: engineering plants and enzymes for biofuels production. Science 2007, 315: 804-807. 10.1126/science.1137016View ArticleGoogle Scholar
- Zhang YH, Ding SY, Mielenz JR, Cui JB, Elander RT, Laser M, Himmel ME, McMillan JR, Lynd LR: Fractionating recalcitrant lignocellulose at modest reaction conditions. Biotechnol Bioeng 2007, 97: 214-223. 10.1002/bit.21386View ArticleGoogle Scholar
- Baker AA, Helbert W, Sugiyama J, Miles MJ: New Insight into Cellulose Structure By Atomic Force Microscopy Shows the Iα Crystal Phase at Near-Atomic Resolution. Biophys J 2000, 79: 1139-1145. 10.1016/S0006-3495(00)76367-3View ArticleGoogle Scholar
- Nishiyama Y, Sugiyama J, Chanzy H, Langan P: Crystal structure and hydrogen bonding system in cellulose Iα, from synchrotron X-ray and neutron fiber diffraction. J Am Chem Soc 2003, 125: 14300-14306. 10.1021/ja037055wView ArticleGoogle Scholar
- Kirby AR, Gunning AP, Waldron KW, Morris VJ, Ng A: Visualization of plant cell walls by atomic force microscopy. Biophys J 1996, 70: 1138-1143. 10.1016/S0006-3495(96)79708-4View ArticleGoogle Scholar
- Davis LM, Harris PJ: Atomic force microscopy of microfibrils in primary cell walls. Planta 2003, 217: 283-289.Google Scholar
- Novotny L, Hecht B: Principles of Nano-Optics. New York: Cambridge University Press; 2006.View ArticleGoogle Scholar
- Veerman JA, Otter AM, Kuipers L, van Hulst NF: High definition aperture probes for near-field optical microscopy fabricated by focused ion beam milling. Appl Phys Lett 1998, 72: 3115. 10.1063/1.121564View ArticleGoogle Scholar
- Stöckle RM, Suh YD, Deckert V, Zenobi R: Nanoscale chemical analysis by tip-enhanced Raman spectroscopy. Chem Phys Lett 2000, 318: 131-136. 10.1016/S0009-2614(99)01451-7View ArticleGoogle Scholar
- Hayazawa N, Inouye Y, Sekkat Z, Kawata S: Near-field Raman imaging of organic molecules by an apertureless metallic probe scanning optical microscope. J Chem Phys 2002, 117: 1296-1301. 10.1063/1.1485731View ArticleGoogle Scholar
- Fragola A, Aigouy L, Mignotte PY, Formanek F, De Wilde Y: Apertureless scanning near-field fluorescence microscopy in liquids. Ultramicroscopy 2004, 101: 47-54. 10.1016/j.ultramic.2004.04.002View ArticleGoogle Scholar
- Nieman LT, Krampert GM, Martinez RE: An apertureless near-field scanning optical microscope and its application to surface-enhanced Raman spectroscopy and multiphoton fluorescence imaging. Rev Sci Instrum 2001, 72: 1691-1699. 10.1063/1.1347975View ArticleGoogle Scholar
- Sánchez EJ, Novotny L, Xie XS: Near-field fluorescence microscopy based on two-photon excitation with metal tips. Phys Rev Lett 1999, 82: 4014-4017. 10.1103/PhysRevLett.82.4014View ArticleGoogle Scholar
- Cheng J-X, Jia YK, Zheng G, Xie XS: Laser scanning coherent anti-Stokes Raman scattering microscopy and applications to cell biology. Biophys J 2002, 83: 502-509. 10.1016/S0006-3495(02)75186-2View ArticleGoogle Scholar
- Nan X, Yang WY, Xie S: CARS microscopy: lights up lipids in living cells. Biophotonics Int 2004, 11: 44-48.Google Scholar
- Ichimura T, Hayazawa N, Hashimoto M, Inouye Y, Kawata S: Tip-enhanced coherent anti-Stokes Raman scattering for vibrational nanoimaging. Phys Rev Lett 2004, 94: 220801-1. 10.1103/PhysRevLett.92.220801View ArticleGoogle Scholar
- Kawata S, Ichimura T, Hayazawa N, Inouye Y, Hashimoto M: Tip-enhanced near-field CARS microscopy. J Nonlinear Opt Phys Mater 2004, 13: 593-599. 10.1142/S0218863504002341View ArticleGoogle Scholar
- Freudiger CW, Wei M, Saar BG, Lu S, Holtom GR, He C, Tsai JC, Kang JX, Xie XS: Label-free biomedical imaging with high sensitivity by stimulated Raman scattering microscopy. Science 2008, 322: 1857-1861. 10.1126/science.1165758View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.